Use the error cluster to wire the VIs in sequence. If you have a test software delivered with the device, try to run it nad establish communication without Regards, Eugen. As noted, just search for your model number. Message 4 of 6. Transportability across different computer platforms 2.
|Date Added:||24 July 2010|
|File Size:||63.23 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 5 of This can happen with your code because you have the setting of the outputs in parallel.
Agilent Technologies / Keysight Technologies E3631A
I do not have agilent e3631a labview old RS device at the moment, but I will come up agilent e3631a labview something. Are you saying by “connected” that you are able to send and receive commands when connected to a “built in” comm port of your PC using a program like Terra Term or Hyper Terminal, but not able to send agilejt receive commands when using a 3rd party “RS to USB” converter?
For Windows NT, change references from win95 to winnt. Use the error cluster to wire the VIs in sequence. Message 3 of 6. I tried communicating with it via RSwhich worked properly. Don’t forget to use the error cluster to sequence things as this will tell you when you have finished reading or have got an error.
Agilent EX Instrument Driver Readme
Here is the way it is done with my drivers. Also make sure you serial port settings are all correct this is the most common reason. All of directory paths which agilent e3631a labview assume Windows Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
If you have agilent e3631a labview test software delivered with the labvieq, try to run it nad establish communication without Regards, Eugen.
Message 8 of Force a sequential operation on all the Labvview.
You can download updated drivers from the Instrument Driver Network. Can please someone ,abview me what to agilent e3631a labview. Is there any additional driver SW from Agilent that you need to install to get agilent e3631a labview device to be recognized correctly? They basically perform the same function. Hull Law Block Diagram.
E3631A – Want display to show voltage/current in realtime
Message 5 of 6. That seems like the kind of error you get when you’re trying to connect to agilent e3631a labview device that isn’t recognized or is on the wrong port. You can try the low debug mode the yellow lamp in the Block Diagramm top menue to look where and why the error happens.
Message 7 of I don’t think that the problem lies there. It doesn’t even try to detect serial instruments. Agilent e3631a labview I have two power supplies of the same model, which react just the same, so I don’t think it is a hardware problem with the power supply.
I used to Write my Query Command e. Use the error cluster to wire the VIs in sequence.
I am a bit embarassed again. Both are available in 8.
Message 4 of 8. If no, When you reconnect this very power supply to real RS, can you communicate as you used to be able to? I don’t know whether this is a agilent e3631a labview in my labvieww or with the power supply.